Tonom GmbH © 2017 All rights reserved.
You are now on the site for health care professionals. Please consider that all the information about Diaton Tonometer is provided for Professionals!
Together Open New Opportunity for Medicine
COMPARATIVE AGREEMENT AMONG THREE METHODS OF TONOMETRY:GOLDMANN APPLANATION,TRANSPALPEBRAL AND DYNAMIC CONTOUR
LUIS A. ZARATE,Magdalena García-Huerta,Rafael Castañeda Diez,Mauricio Turati,Felix Gil Carrasco,Jesus Jimenez-Roman,Jose A. Paczka.1GLAUCOMA,ASOCIACION PARA EVITAR LA CEGUERA EN MEXICO,Mexico;2.Glaucoma;3.Asistencia e Investigacion en Glaucoma
Purpose: To investigate agreement of intraocular pressure (IOP) as measured by the Goldmann applanation tonometer (GAT), the Pascal dynamic contour tonometer (DCT), and Diaton transpalpebral tonometer (DTT).
Methods: Device agreement was calculated by Bland-Altman analysis in 77 eyes of 40 individuals (mean age 58.9 ± 13 years) with a mixed diagnosis of glaucoma suspicion and primary open-angle glaucoma. All measurements were performed in a random order by the same clinician according to standard procedures.
Results: Mean IOPs ± S.D. were 14.4 ± 2.9 mm Hg (GAT), 18.8 ± 3.2 mm Hg (DCT; P = 0.005, ANOVA), and 15.1 ± 3.1 mm Hg (DTT). Bland-Altman analysis demonstrated that, on average, DCT IOP measurements overestimated in approximately 3 mm Hg, values derived from GAT and DTT, although agreement was fairly good.
Conclusions: All methods of tonometry were adequate to measure IOP in our sample. Agreement among devices was considered good; nevertheless, DCT values of IOP were significantly higher as compared to the other two assessed methods.